SGS 40,000Hz 180W Ultrasonic Bath Cleaner
|Place of Origin:||CHINA|
Payment & Shipping Terms:
|Minimum Order Quantity:||1 pcs|
|Packaging Details:||1unit/carton box|
|Delivery Time:||3-5 working days|
|Payment Terms:||Western Union, T/T|
|Supply Ability:||1000PCS /DAY|
|Certification::||CE, RoHS||Heating Power:||200W|
|Material:||SUS304||Product Name:||4..5L 030S Tabletop Ultrasonic Cleaner|
40 000Hz Ultrasonic Bath Cleaner,
SGS Ultrasonic Bath Cleaner,
SGS ultrasonic bath sonicator
Durable bench top Ultrasonic Cleaner 4.5L 180W For surgical instruments leb Musical Instruments
|Material of tank||SUS304|
|Material of shell||SUS304|
|Timer||0~30 mins adjustable|
|Power supply type 1||AC 100~120V, 50/60Hz|
|Power supply type 2||AC 220~240V, 50/60Hz|
|Tank inner dimension||300*150*100mm|
|Inner Packing size||410*255*320mm|
|Certificate||SGS & FCC & CE & RoHS|
Similar models with different capacity:
|Model||Capacity||Tank Size||Ultrasonic Power||Heating Power||Frequency|
Suggested Ultrasonic Cleaning Procedure
In all cases manufacturers’ instructions should be followed when using an ultrasonic cleaning process. These are representative steps.
Fill the ultrasonic cleaning tank with an approved medical instrument cleaning solution such as CLN-LR012 available from Tovatech following dilution instructions provided. Turn the cleaner on to start the degassing process. This step removes entrained air in new solutions that interferes with the efficiency of cavitation and takes approximately 10 minutes.
In the meantime:
- Segregate instruments by alloy or composition to avoid potential damage (Chromium plated instruments should not be cleaned ultrasonically)
- Instruments with movable parts should be disassembled to facilitate cleaning
- Place the instruments the ultrasonic cleaner’s mesh basket, taking care that they do not come in contact with each other
- Cannulated or lumened instruments should be positioned to insure interiors are wetted with the cleaning solution. In some instances placing them on an angle will facilitate this
- Set the control panel per manufacturers’ instructions and start the cleaning process
At the end of the cycle, remove the instruments from the ultrasonic cleaning bath and thoroughly rinse them to remove all traces of the cleaning solution. Deionized water rinses will avoid spotting. If the instruments are not to be immediately disinfected and sterilized be certain that they are thoroughly dried and protected. Part reassembly can occur after sterilization.
Procedures should be in place to guide the replacement of used ultrasonic cleaning solutions. In some instances it is recommended that solution be drained and tanks thoroughly cleaned and dried after each ultrasonic cleaning cycle. Most solutions available today are biodegradable, which facilitates disposal but local authorities should be consulted on proper practices.
Jewelry industry, medical industry, electronic factory, molding factory, car workshop, diesel workshop, car industry, scientific laboratory, university, dental clinics, eyeglass shop, hardware tools shop, printing industry.